|Eric Wild 20d96b5d74|
This utility uses a UHD device with timestamp support to measure the latency between transmitting a pulse and receiving it back.
You can measure the apparent latency of the device itself (internal processing, analog delays, ...) by using a loopback. Or you can also measure the latency of an external device. Make sure to first make a measure of the UHD device own latency and substract that from you measurement to get the real latency of the DUT.
This latency is a function of the sample rate and the master clock rate, so make sure to set those parameters correctly !
How it works
This repeatadely transmits a burst made of QPSK symbol, filtered by a simple RRC.
Ideally instead of a random LFSR, this should use codes that have an auto-correlation as close as possible to the delta function, but this is not implemented yet.
At the same time a transmit burst is sent, the UHD device is asked to receive some samples at the same timestamp. Then in those receive samples, we look for the same burst we transmitted (using correlation) and display the position (in samples) that this was found in.
The correlation can have false 'echos' and sometimes needs to be interpreted rather than used as raw data directly.
The receive window is limited to a much smaller length than the pulse period to make sure the host has time to receive the data, correlate it before sending the next pulse. (This isn't especially optimized ATM).
If you see 'RX Stall' errors, try increasing the burst period, or diminishing the receive window ('Max delay').
Starting the utility with default options should show something like :
$ ./pinger [+] Options : . TX frequency : 1000.000 MHz . TX gain : 60.0 dB . RX frequency : 1000.000 MHz . RX gain : 60.0 dB . Master Clock Rate : Auto . Sample Rate : 2.000 Msps . Burst length : 256 samples . Burst period : 250.000 ms . Maximum delay : 5.000 ms
This means that it will transmit a 256 sample long pulse every 250 ms. The receive window is 5 ms..
Typical output would be something like :
[INFO] [UHD] linux; GNU C++ version 6.4.0; Boost_106500; UHD_18.104.22.168-99-g8e7768c7 [INFO] [B200] Loading firmware image: /opt/gnuradio-37qt5/share/uhd/images/usrp_b200_fw.hex... [INFO] [B200] Detected Device: B205mini [INFO] [B200] Loading FPGA image: /opt/gnuradio-37qt5/share/uhd/images/usrp_b205mini_fpga.bin... [INFO] [B200] Operating over USB 3. [INFO] [B200] Initialize CODEC control... [INFO] [B200] Initialize Radio control... [INFO] [B200] Performing register loopback test... [INFO] [B200] Register loopback test passed [INFO] [B200] Setting master clock rate selection to 'automatic'. [INFO] [B200] Asking for clock rate 16.000000 MHz... [INFO] [B200] Actually got clock rate 16.000000 MHz. [INFO] [B200] Asking for clock rate 32.000000 MHz... [INFO] [B200] Actually got clock rate 32.000000 MHz. [+] Echo at : [+] Echo at : 15 (71.541344) [+] Echo at : 50 (1.466832) [+] Echo at : 50 (1.351750) [+] Echo at : 50 (1.278904) [+] Echo at : 50 (1.423250) [+] Echo at : 50 (1.461555) [+] Echo at : 50 (1.621279) [+] Echo at : 50 (1.626796)
This means that for those parameters, we see our TX pulse echoed back on RX 50 samples later.
Obviously this isn't actual RF delay, this is just a mismatch of UHD TX and RX timestamps in this case.